What is bottle washer?
Bottle washer definition
is a widely used cleaning equipment, bottle washer is mainly a special equipment for cleaning the inside and outside of the bottle or other similar bottles in the mechanical continuous transport. At present, no matter what kind of bottle washing machine most of the use of ultrasonic cleaning technology and ultrasonic cavitation effect of mechanical friction, in order to clean with the general bottle washer difficult to clean the bottle inside and outside the more firmly adhered to the material.
Laboratory bottle washer as one of the instruments that began to popularize the purchase of laboratory in the past 10 years, mainly through the control of water temperature, detergent emulsification stripping effect, strong water scouring force, to work together to achieve the purpose of cleaning glassware, cleaning uniformity, cleaning results can be controlled, cleaning automation for its advantages.
Application of bottle washer
are used in a wide range of laboratories in organic, inorganic, physical chemistry, biology, microbiology, medical, pharmaceutical, food or cosmetic industries. The different fields of application are also important influencing factors in determining the type of machine and accessories needed as well as the type of cleaning procedure and cleaning agent.
Bottle washer working principle
The specific working principle of the bottle washing machine is as follows,
In the bottle washing machine, the unpacked bottles are transported to the bottle feeding screw through the conveying mesh belt, and then divided into a series of star-shaped and navigated, and the bottles are transferred to the descending bottle seal, during which the bottles are filled one by one at the filling table, and the descending bottle seal is driven down by the bottle-holding table, under the impetus of the super washing screw, the bottles are passed through the ultrasonic excitation area in turn, and then driven by the bottle-holding table at the ascending bottle seal. Then the bottle will be driven by the rising bottle sealing table, and the bottle neck will be caught by the bottle clamping machine, and the bottle clamping machine will turn the bottle upside down under the driving of the big turntable, and enter into the bottle washing process which is arranged in a circle.
* Washing inside and outside with circulating water.
* Washing inside and outside with circulating water again.
* Compressed air forced water.
* Injection water cleaning.
* Compressed air cleaning residual water.
* Compressed air cleaning inside and outside again.
Then, the bottle clamping robot flips again, and the bottle returns to the main state, and is discharged to the next drying process.
Composition of bottle washer
Glass bottle washer mainly consists of the following parts,
400-500 liters per minute circulation pump provides extremely high cleaning capacity.
Has standard spray arms for upper and lower layers.
Purified water feed system to inject fresh, clean purified water prior to each cycle. The system is directly connected to hot water, cold water or purified water pipes. A circulation speed sensor ensures that the pump is running at the correct speed. If an obstruction is detected, the pump will be shut down before it overheats to prevent damage to the machine.
The total heating power is 4KW-5KW watts, no matter how many degrees the incoming water temperature is, it can ensure that the appropriate washing and rinsing temperature can be reached quickly.
The temperature of the purified water for washing and rinsing can be adjusted to a maximum of 93℃.
The built-in imported cleaning agent is dispensed with the corresponding dose through the peristaltic pump, and the glassware is cleaned through the cleaning system.
The liquid acid neutralizer can be dosed using the built-in pump. The neutralization step helps to ensure the removal of all residual cleaning agents.
The four-stage effluent filtration system protects internal components by utilizing surface, coarse, finer and fine filters to prevent debris from entering the recirculation during the cleaning process.
A mesh filter in the inlet pipe provides additional protection.
Independent air heater with high-efficiency circulation fan to improve the drying efficiency of the vessels. The drying temperature of 40~90℃ can be controlled.
Adopt a HEPA high-efficiency air filter to ensure the cleanliness of the air entering the drying system and avoid the secondary contamination of the vessels by air.
Laboratory bottle washer drying hot air through the nozzles of the top and bottom spray arms as well as the nozzles of the cleaning column to ensure rapid drying of glassware inside and outside.
Benefits of using bottle washer
1. Bottle washer can standardize the cleaning process to ensure a uniform cleaning effect and reduce the uncertainty arising from the human operation.
2. Reduce the risk of staff and avoid injury or infection caused by personnel in the manual cleaning process.
3. Washing, disinfection, and drying are done fully automatically, reducing equipment and labor input, and saving water resources and costs for users.
4. Cleaning quality meets GMP specification requirements.
5. Multi-specification and multi-purpose can be used in linkage or single machine.
Types of bottle washer
Turntable bottle washer
Turntable bottle washer is a semi-mechanical, semi-manual type of bottle washer, the model is backward, and only a few hospital preparation rooms are still in use.
Crawler bottle washer
Crawler bottle washing machine adopts the gap movement of the crawler, and there are two pairs of pneumatic manipulators in front and back to load and unload the glass bottles so that the glass bottles are always facing down in each station of washing. The bottle mouth always faces down in each station of washing, to facilitate the clean effect of bottle washing. The disadvantage is that coarse and fine washing can not be isolated, the production capacity is low, and there is a certain degree of difficulty in debugging, which has been gradually eliminated.
Drum bottle washer
Drum type bottle washer is characterized by mature bottle washing process, and applicable to acid and alkali washing two kinds of process. The bottle washing effect is more reliable, the machine result is simple, easy to manufacture and debug, coarse and fine washing can be separated, it is a prototype of large crawler bottle tray type bottle washing machine. Its shortcomings are the strict requirements for glass bottle geometry, followed by a trace of residual liquid in the glass bottle after each washing process.
Chamber type different end in/out bottle washer
The high degree of automation of the chamber type different end in/out bottle washing machine integrates the washing functions of lye, hot water, brush, distilled water, etc., and can automatically and continuously carry out bottle washing operation. The machine is combined with the requirements of GMP for infusion production so that the incoming and outgoing bottles are carried out at the two ends of the compartment respectively to prevent cross-contamination.
Bottle washer specifications and features
Stainless steel bottle washer
|Inner cavity size||688*700*703mm
|Water circulation flow(L/min)||1000
|Water heating power(KW)||5
Type: Chamber type different end in/out bottle washer
1. Stainless steel bottle washer designed to arrange the cleaning bits according to the principle of fluid mechanics to ensure the cleanliness of each item cleaned.
2. Rotating spraying arm with optimally designed flat-mouth type nozzle to ensure 360° dead-end coverage by spraying.
3. Cleaning column side oblique incision to ensure 360° cleaning coverage of the inner wall of the vessel.
4. Height adjustable bracket to ensure effective cleaning of different sizes of utensils.
High-pressure bottle washer
|Inner cavity size||600*629*635
|Water circulation flow(L/min)||800
|Water heating power(KW)||5
Type: Chamber type different end in/out bottle washer
1. High-pressure bottle washer adopts double water temperature control to ensure the whole cleaning water temperature.
2. Washing liquid can be set and added automatically.
3. Built-in HEPA filter to ensure the cleanliness of the dry air.
4. According to different cleaning needs, users can customize the combination of cleaning programs.
Cleaning solution for bottle washer
Glassware cleaning is an important part of the laboratory work, in the bottle washer work cleaning agent is essential, the following is an introduction to the commonly used in the laboratory bottle washer cleaning solution
.Industrial concentrated hydrochloric acid
Technically concentrated hydrochloric acid is produced by burning chlorine gas in hydrogen gas in a preservative-treated cylinder to produce hydrogen chloride, which is then passed into water. Industrial concentrated hydrochloric acid can remove scale and some inorganic salt precipitation.30% nitric acid solution
Mostly used for washing micro-dropper and carbon dioxide meter.5% oxalic acid solution
Acidified with several drops of sulfuric acid, can effectively remove potassium permanganate residue.5-10% trisodium phosphate (Na3PO4- 12H2O) solution
Can effectively wash oil stains.Sodium hydroxide solution containing potassium permanganate and ethanol solution of potassium hydroxide
These two strong alkaline washing solutions can effectively remove dirt from the inner walls of glassware. However, because of the strong erosive nature of glassware, care should be taken to control the washing time should not be too long.5% to 10% disodium EDTA (EDTANa2) solution
The solution is heated and boiled and used to clean the white precipitates on the inner wall of glass instruments.Urea detergent solution
Urea washing solution is a good solvent for protein, mostly used for washing residual protein glassware.Organic solvents
Common organic solvents
such as xylene can clean paint, ethanol can clean grease, and fat-soluble statins.Chromic acid washing solution
Add 100mL of concentrated sulfuric acid to the beaker and slowly add 5g of potassium dichromate powder while heating the beaker, stirring continuously during the addition. After all the potassium dichromate powder is dissolved and cooled, a chromic acid washing solution is produced. This washing solution can effectively remove oil, handprints, and other dirt.
Precautions for using bottle washer
When using the glass bottle washer, do not let the water pipe touch chemical reagents or other hot objects. Only suitable utensils should be put into the bottle washer for cleaning. Do not use any other kind of solvent in the bottle washing machine. Do not open the door while the machine is running, hot steam may cause injury to personnel. The power must be turned off and the water inlet valve closed during maintenance or long periods of non-use. To avoid danger, replace the power cord by the manufacturer's professional technicians.
How to clean bottle washer?
Before cleaning the bottle washer, make sure to cut off the power supply, otherwise there are certain safety hazards. When cleaning the exterior of the instrument or rubber parts, use a soft cloth to wipe, and avoid using products with abrasives or solvents. When cleaning the inside of the instrument, if the stain or spot is difficult to remove, use a soft cloth with an appropriate amount of white vinegar dilution to wipe. When cleaning the control panel of the instrument, to avoid abrasion, do not use rough or sharp cleaning products to wipe. It is recommended to choose a soft cloth with an appropriate amount of electrical polish for cleaning, and let it dry thoroughly.
The filter can effectively separate water and solids to prevent the water circulation system from being damaged. Therefore, to ensure the good performance and cleaning effect of the instrument, the filter should be cleaned regularly. It is recommended to remove the filter after the glassware is cleaned, pour off the solids, and rinse it with tap water. After the instrument has been cleaned, disconnect the water source and open the bin door for ventilation to avoid moisture or odors inside the instrument.
How to maintain bottle washer?
When transporting or moving the glass bottle washer, keep it horizontal and avoid excessive tilting or being bumped. To ensure the normal operation of the instrument, please check and replace the parts that rotate frequently, such as hoses, peristaltic pumps, circulation pumps, etc. regularly. If the instrument is not used for a long time, it is recommended to run the "Quick Clean" program once every half month to thoroughly clean the pipeline and internal cavity. After the program, please cut off the power, close the water inlet valve, and open the door to dry the cleaning compartment.
How to buy bottle washer?
ANTITECK provide lab equipment, lab consumable, manufacturing equipment in life sciences sector.
If you are interested in our bottle washer or have any questions, please write an e-mail to email@example.com, we will reply to you as soon as possible.